View and Download Hach Sc E instrument manual online. Analysis System. Sc E Measuring Instruments pdf manual download. Hach Sc Controller User Manual – Download as PDF File .pdf), Text File .txt) or read online. Hach Sc Controller User Manual . Hach Company, – .. 37 W with 25 W sensor/network card load 24 VDCpowered sc model: 24 VDC.
|Published (Last):||12 October 2018|
|PDF File Size:||5.83 Mb|
|ePub File Size:||20.17 Mb|
|Price:||Free* [*Free Regsitration Required]|
Table of Contents Section 5 Maintenance System Startup Two types of verifications are offered. Havh Dimension Illustrations Installation 3.
Depends on sample Clean the sensor characteristics. Mounting foot for panel mounting 2Cat. Mounting bracket, controller, Cat.
Enter text from picture: Appendix A Modbus Protocol A.
In the United States: Page 21 Installation 3. Controller Display Features Operation 4. All configuration settings and calibrations can be changed.
Wet Verification System Startup 5. Section 5 System Startup 5. Section 2 General Information 2. Section 4 Operation Section 4 Operation 4. Page 6 Table of Contents Section 5 Maintenance Maximum cable length is 9. Twist the bulb in a counterclockwise direction, pulling out slightly, until it is released from the housing. Pay attention mmanual all danger and caution statements.
HACH SC E INSTRUMENT MANUAL Pdf Download.
Section 7 Troubleshooting Section 7 Troubleshooting 7. This is the default setting. To print the manual completely, please, download it. Page 12 Installation 3.
The controller display shows the current reading plus a secondary measurement such as temperature if connected to a single sensor, or two readings with their corresponding secondary measurement readings when two sensors are connected. Table 1 Turbidimeter Hxch Nephelometric light scatter at 90 degrees relative to the incident light beam.
Flat washer 4Cat. Page 20 Installation 3.
Hach sc100 1720E Instrument Manual
Wear cotton gloves to protect your hands and to avoid fingerprints on the bulb. Measurement Xfer Transfer 4. Section 7 Warranty, liability and complaints HACH LANGE GmbH warrants that the product supplied is free of material and manufacturing defects and undertakes the obligation to repair or replace any defective parts at zero cost.
Pass the cable through conduit and a conduit hub or a strain relief fitting and an available access hole in the controller enclosure.
The frequency will depend on the nature and concentration of dissolved and suspended nanual in the sample. Inadequate bubble removal from Increase the signal averaging time to a longer interval.
Table of Contents S1c00 1 Specifications Pull the lamp leads and connector through the hole in the lamp housing. Printed in the U. Instrument Setup Operation 4.
Hach-Lange SC : Owner manual
Page 32 Operation 4. Close and secure the cover. Periodic inspection of these devices is recommended to check for degradation.
The following two options are available: The passcode sc00 factory set to sc the five digits must be followed by a space to remove the trailing asterisk. The keypad consists of the eight keys described in Table 7. Close and secure the cover.
In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase price, subject to the pro-rated schedule above, excluding original shipping and handling charges.
Section 6 Replacement Parts 6. Page 71 Index Sample Line Installation DEADBAND—Sets the range where the relay remains on after the measured value decreases below the setpoint value high phase relay or increases above the setpoint value low phase relay.